(RadoInspector Master Device) program is designed to control the hardware instruments (spectrum analyzers, measuring and scanning receivers, GPS receivers) using physical data transfer interfaces and protocols (COM-port, IEEE488, LAN, USB). The RadioInspectorRC
programs (if connecting to instruments on a remote computer) use the I_MasterDevice program on the remote computer to control the remote instruments.
program must be installed on the same computer where RadioInspectorRC
is running (the RadioInspectorRT program can control its own devices connected to the local computer) or I_MasterDevice must be installed on a remote computer that is connected through a network (LAN or Internet) to the host computer where RadioInspectorRC
programs are run.
How it works: RadioInspectorRC
controls devices on a remote computer) programs send a request through the network to the I_MasterDevice
program and receive a response with results. Requests can be represented as: "Execute a measurement of a spectrum in the frequency span S, with a central frequency F, bandwidth B, video bandwidth WB, detector D, reference level R…". In response the I_MasterDevice program configures the relevant instrument settings and the instrument executes measurements. The results (an array of measurement results) are sent as response to this request.
If the program is performing continuous measurements (for instance, if the "Start" button in RadioInspectorRC
is pressed), the instrument is reserved to the exclusive use for the RadioInspectorRC
program. Thus, the I_MasterDevice
program implements a function of control and allocation of measuring instrument's resources. I_MasterDevice
operates a GPS receiver (if connected) and adds coordinates to each measurement result. This enables mobile monitoring and analysis linking measurement results to geographical coordinates for processing and integration with digital maps.
program can determine calibration data for the antenna in use and the calibration data of intermediate transmission paths (antenna switches, feed line loss, etc.) for inclusion in field strength measurements.
with RemoteMD option